The Synergy HT Multi-detection reader is a robotic-compatible microplate .. and J. Sambrook () Molecular Cloning A Laboratory Manual, Cold Spring. View and Download BioTek Synergy HT operator’s manual online. Multi-Mode Microplate Reader. Synergy HT Laboratory Equipment pdf manual download. Manuals and User Guides for BioTek Synergy HT. We have 2 BioTek Synergy HT manuals available for free PDF download: Operator’s Manual.
|Country:||Republic of Macedonia|
|Published (Last):||3 October 2010|
|PDF File Size:||6.10 Mb|
|ePub File Size:||7.7 Mb|
|Price:||Free* [*Free Regsitration Required]|
The instrument should be serviced by BioTek authorized service personnel.
Procedure For Models Without Injectors Page Z position. Instrument Qualification Chapter 4 Instrument Qualification This chapter contains procedures for qualifying the initial and ongoing performance of the Synergy HT and the external dispense module if used. Remove the Shipping Panel Figure 2: The calibration sticker is affixed directly to the plate. Scenario 4 fluorescence autocalibration: Page 4 BioTek Instruments, Inc. Running A Dispense Protocol optional Chapter 5: If leaks are detected, tighten all fittings and repeat the prime.
BIOTEK SYNERGY HT OPERATOR’S MANUAL Pdf Download.
Absorbance Liquid Test 1 Bold text is primarily used for emphasis. See page 6 for contact information. Microplates should be perfectly clean and free of dust or bottom scratches. If your assays use incubation, we recommend enabling Temperature Control and synerg the incubator to reach its set point before running the System Test.
Error Codes Fatal Errors Fatal errors indicate conditions that require immediate attention.
If you have not already done so, save the file with an identifying name. If sodium hypochlorite bleach was used, perform Rinse the Fluid Lines on the next page. See the photos on the next page.
This error can apply to either Syringe 1 or Syringe 2. Clean The Priming Plate Clean the Internal Components Stand facing the aynergy of the instrument.
One end of the cable connected to the port on Page This test is performed when the instrument is first turned on and then tested periodically during background functions.
Recommended Qualification Schedule The fluid should pump through the tubing and dispense into the priming plate. Synerg Package Contents Part numbers and package contents are subject to change. COM port to which the reader is connected.
BioTek Synergy HT Manuals
Place the tip priming trough in the left-rear pocket of the carrier. Models with injectors bioteo Repeat steps 4—9 using the Synergy HT Dispenser 2.
Turn off and unplug the instrument for all decontamination and cleaning operations. This frequency may be adjusted depending on the trends observed over time.
BioTek Synergy HT Operator’s Manual
Install The Fluorescence Lamp Assembly Plug the lamp biotel into the power source located to the right of the lamp. Test Solution Recipes Chapter 4 Instrument Qualification This chapter contains procedures for qualifying the initial and ongoing performance of the Synergy HT and the external dispense module if used. Establish Communication COM port to which the reader is connected. Chapter 5 Preventive Maintenance This chapter provides step-by-step instructions for maintaining the Synergy HT and external dispense module if used in top condition, to ensure that they continue to perform to specification.
The Synergy HT is completely computer-controlled; it is not equipped with a keypad or its own user interface. Comments Click Start Test. Clean Exposed Surfaces Models With Injectors 0. An alternate method that may be used to determine accuracy, linearity, and repeatability is Liquid Test 2, described on page